Development and Validation of Stability Indicating RP-HPLC Method for Rivaroxaban and Its Impurities

Rivaroxaban is oxazolidinone derivative having anticoagulant activity. In literature few analytical methods are discuss about estimation of rivaroxaban; but rarer discussion is available for rivaroxaban impurity profile. The objective of this study is to develop and validate RP-HPLC method for the qualitative analysis of Rivaroxaban. The chromatographic separation was achieved on ZorbaxSB C18 (250 mm X 4.6 mm, 3.5 μ) HPLC column using buffer (0.02M mono basic potassium di hydrogen phosphate) and solvent mixture (acetonitrile: methanol mixture) ingradient programme. The developed methods were validated as per ICH guideline and found to be specific, precise, sensitive and robust.


Introduction
Rivaroxabanis5-chloro-N-({(5S)-2-oxo-3-[4-(3oxomorpholin-4-yl)phenyl]-1,3-oxazolidin-5yl}methyl) thiophene-2-carbinoxamide Figure 1.Rivaroxaban is an oxazolidinone derivative anticoagulant that competitive reversible antagonist of activated factor X (Xa).Factor Xa is the active component of the prothrombinase complex that catalyses conversion of prothrombin (factor II) to thrombin (factor IIa).It is a highly selective direct Factor Xa inhibitor with oral bioavailability and rapid onset of action.Rivaroxabin does not inhibit thrombin (activated Factor II), and no effect on platelets have been demonstrated In November 2008 the Therapeutic Goods Administration approved new oral anticoagulant drug Rivaroxaban for the prevention of venous thrombosis in patients having elective knee or hip replacement [1,2].
There is no official monograph available for Rivaroxaban or drug product in any pharmacopiea.A preliminary survey of literature for suitable method development for Rivaroxaban has been made [3,4].Review of literature suggests that no extensive work has been carried out for the routine analysis of Rivaroxaban, Which can address all process impurities and degradation profiles [5][6][7].Monitoring of impurity profiling is very important for quality of drug and patient safety purpose.Also literature survey shows few analytical methods were published for the estimation of Rivaroxaban during formulation and bio availability study for the assay purpose.But rare discussion is available for Rivaroxaban impurity profile study.This study shows detail discussion on monitoring of commercial rout of synthesis and impurity profiling Figure 2, 3. Hence the aim of the present work was to develop accurate and robust routine HPLC method.

Experimental Material and Reagents
Pure Rivaroxaban was obtained using commercial route of synthesis as per the process described in the Figure -2 [9].The related impurities including process impurities and degradant impurities (as described in Figure -3) were synthesised in house.The rivaroxaban standard and impurities were characterized using proton nuclear magnetic resonance and mass spectrometry equipped with HPLC.HPLC grade acetonitrile, methanol was procured from J T Baker.Analytical grade potassium dihydrogen phosphate and orthophosphoric acid obtained from Merck chemicals.HPLC grade water obtained from Millipore system was used throughout the analysis.Ion pair reagent Octane sulfonic acid was purchased of Ranchem make.

Development and Validation of Stability Indicating RP-HPLC Method for Rivaroxaban and Its Impurities
Copyright: © 2018 Hanana, et al.In the process of developing RPHPLC method three key parameters were studied which influence the selectivity such as chemistry of stationery phase, pH of the buffer, and organic modifiers.Phophate buffer with Octane sulphonicacid, pH was screened as pH3.0,4.0 and 5.0.HPLC columns were used for development of method were Inertsil ODS 3V, Zorbax phenyl, Kromasil C18 and ZorbaxSBC18.Methanol and Acetonitrile were chosen individual and in different ratio as organic modifier.The impurities spiked solution in Rivaroxaban and sample of all stressed condition were studied and recorded and method was optimized with satisfactory resolutions among all impurities on ZorbaxSBC18 column in Figure 4.

Instrumentation and Chromatographic Conditions
Agilent HPLC 1200 (Agilent Technologies, Germany) equipped with photodiode array detector was used for method development, forced degradation studies and method validation.Zorbax SB C18 (250mmX4.6mm 3.5 µ) HPLC column.Column thermostat at 45˚C was used for the impurities separation.Buffer was prepared using 0.02M of anhydrous potassium dihydrogen phosphate and 1 gm of Octane sulphonic acid solution was adjusted to pH 3.0 with orthophosphoricacid.Diluent was prepared by mixing solution A and water in the ratio of 500:500 v/v.Solution A was prepared by mixing methanol and Acetonitrile in the ratio of 500:500 v/v.The test sample solution having concentration of 1000 µg/ml was prepared for the determination of related substances.The stock solution was prepared by dissolving each impurity (Impurity1, Impurity2, Impurity3, Impurity4, Impurity5, Impurity6, Impurity7, Impurity8, Impurity9 and Impurity10) at concentration about 15 µg/mL in diluent and further diluted up to 1.5µg/ml along with Rivaroxaban standard at 1000 µg/ml to prepare the system suitable solution.Inject diluted standard solution in six replicates into the chromatograph and record the chromatogram% RSD for area response of Rivaroxaban peak from six replicate injections of diluted standard solution should not be more than 5.0.The blank, system suitability solution and sample solution of 1000µg/ml, were injected separately and chromatographed under the optimized chromatographic conditions.The resolution NLT 2.0, between Impurity 8peak and Impurity 9 were set as system suitability criteria.All impurities were quantified against 0.1% rivaroxaban diluted standard solution applying the derived Relative Response Factor (RRF).The relative retention time with respect to Rivaroxaban peak and RRF of all impurities are as shown in Table-1, 2.

Specificity (Selectivity)
Specificity is the ability of method to measure the analyte in presence of its potential impurities.Stress testing of the drug substance performed to identify likely degradation impurities, which intern help to establish the degradation pathways and intrinsic stability of the molecule and validate the stabilityindicating power of the analytical procedures used [8,9].

The specificity of developed RP-HPLC method for
Rivaroxaban was determined in presence of its impurities (Impurity 1 to Impurity10 ) and degradation products.Forced degradation studies were also performed on Rivaroxaban to provide an indication of the stability-indicating property and specificity of the proposed method.The dry degradation study was performed by exposing the sample to different stress conditions such as light (1.2 million lux hours), heat (80˚C for 12hours), hydrolytic condition (45°C, 75%RH for 48 Hrs.).Wet degradation was performed as acid hydrolysis (1 M HCl for 4hrs at 80°C), base hydrolysis (1 M NaOH for 4hrs at 80°C) and oxidation (5%v/v H 2 O 2 for 4 hrs at 40°C).Rivaroxaban was found to degrade significantly in Acid condition, but impurities are found well separated and found method is specific.Mass balance was observed during degradation for all the stressed samples.

Linearity
Linearity solutions were prepared by quantitative dilutions of the stock solution of impurity standard and main drug standard to obtain solutions at LOQ to 250% of the specification limited.Known impurity at 0.15% level and unknown impurity at 0.1% level.A series of solutions were prepared by quantitative dilutions of the stock solution of main drug to obtain solutions at 80% to 120% of the sample concentration.
Each solution was injected and areas were recorded.The linearity of peak areas versus different concentrations was evaluated for Rivaroxaban and its related impurities.The linear regression data for all the impurities plotted and correlation coefficient for all impurities was above 0.99.Linearity results are shown in Table 3.

Validation parameter
Imp-

Limits of Detection and Quantification (LOD and LOQ)
According to ICH Q2 (R1) recommendations the Limits Of Detection (LOD) and the Limit Of Quantification (LOQ) for Rivaroxaban and its process related impurities (Impurity 1 to Impurity 10 ) were estimated by calibration curve method [standard deviation of the response (σ) and the slope (S)], by injecting the series of dilute solutions of known concentration.The values of LOD and LOQ found are as depicted in Table-4.
Precision was studied at the LOQ level by injecting six individual preparations of Rivaroxaban and its impurities, followed by the calculation of % RSD of the peaks areas.The %RSD of LOQ precision was below 10%.

Precision
The precision of method is degree of agreement between the results.Precision of the method was studied for system precision, method precision and intermediate precision.A standard solution of Rivaroxaban at 0.1% was injected for six time to determine the system precision of the method and %RSD was calculated for Rivaroxaban.The %RSD of system precision was found about 0.69%.Six separate test sample solutions of Rivaroxaban were prepared by spiking the related impurities (Impurity 1 to Impurity 10) at limit level (i.e.0.15% for known and 0.1% for unknown).The % RSD (n = 6) for each related impurities was evaluated and found in between 0.72% to 2.44%.The similar procedure of method precision was carried out by a different analyst, using different mobile phase and diluent preparations and instrument on a different day with different lot of same brand column for intermediate precision study.The %RSD of results for intermediate precision study was calculated and compared with the method precision results.

Accuracy (Recovery)
Accuracy of the method for all the impurities was determined by analyzing Rivaroxaban sample solutions spiked with all the impurities at four different concentration levels of LOQ, 50 %,100 % and 250% of each at the specified limit in both methods.The recovery of all these impurities were found to be in-between the predefined acceptance criterion of 80.0% -120.0%.

Stability of Analytical Solution
Rivaroxaban spiked with all impurities at specified level were prepared and analyzed immediately and after different time intervals up to 24 hrs to determine the stability of sample solution in both methods.The sample cooler temperature was maintained at about 25 0 C and at about refrigerator temperature (2-8 0 C).The results from these studies indicated that the sample solution was stable at room temperature and at 2 -8 0 C.

Robustness
The chromatographic conditions were deliberately altered to evaluate the robustness of developed method.The resolution between closely eluting peaks was evaluated on altered chromatographic condition.To study the effect of flow rate on the resolution, the flow rate of mobile phase was altered by ± 0.1 mL/min (0.9 to 1.1 mL/min from 1.0 mL/min).The effect of column temperature on resolution was studied at 40 0 C and 50 0 C instead of 45 0 C. whereas all other mobile phase components were held constant similarly to study the pH effect, pH of buffer was altered by ± 0.2keeping rest parameters same.Buffer: solvent mixture was studied by changing composition ±2% absolute of solvent mixture.All these parameters were studied by changing one parameter only at a time.The resolution between all known and unknown peaks present in sample was greater than 1.5 in all the deliberate varied chromatographic conditions indicating the robustness of the method.

Conclusion
A simple accurate and precise HPLC method has been developed for determination of Rivaroxaban and its impurities in bulk drug and dosage form.The method was successfully validated in accordance with ICH guidelines.It can be conveniently used for routine quality control analysis of Rivaroxaban and its impurities in bulk drug and dosage form.Degradation impurities not interference with Rivaroxaban and their impurities, thus the method is stability indicating.

Figure 2 :
Figure 2: Commercial rout of synthesis for rivaroxaban

Figure 4 :
Figure 4: All impurities spiked at 0.15%level in Rivaroxaban Solvent mixture was prepared Acetonitrile: Methanol in ratio 820:180v/v.Mobile phase A was prepared by mixing Buffer and solvent mixture in ratio 800 :200 v/v.Mobile phase-B was prepared by mixing Buffer and solvent mixture in ratio of 200:800 v/v.The flow rate and injection volumes were 1.0ml/ min and 10µlrespectively.The analysis was carried out under the gradient condition as time(min)/A(v/v):B (v/v); T 0.01 /85:15, T 22.0 /75:25, T 35.0 /55:45, T 50.0 /40:60 ,T65.0/40:60,T 66.0 /85:15 and T 75.0 /85:15.The data was acquired at 240nm for Impurity 1, to Impurity 10; Run time kept 75 min.Chromatographic data processed by using chemstation and chromline HPLC software.The photodiode array detector was used to determine the peak purity of stressed sample.Preparation of Solutions and Analytical Procedure (System Suitability)

Table - 4
: LOQ and LOD values for Rivaroxaban and its impurity